Applies to: Chromatography Smart (GC/MS) Tickets Solved: 0 Rank: 3.5
The over temp error is common in a Service Module whose oil wick and diaphragm have become deficient. Power off the Service Module and HAPSITE and allow the power to drain from both instruments. Wait approximately one hour before turning on the instruments and attempting to complete the previous operation.
This is a temporary fix; to address this issue permanently, call technical support.
Our SMEs are standing by to assist with issues and repairs.
Was this helpful? Yes No
Keywords: Oil, wick, diaphram, Temp error, Shutdown, Turbo Speed,