Applies to: Radiation Identifinder (Gamma Spectroscopy) Tickets Solved: 0 Rank: 4.5
This particular error while calibrating the IdentiFINDER can be caused by radiation sources in the area, a weak calibration source, or by an extreme change in temperature. To overcome this error remove all sources or radiation and/or try a different cailbration source. If the area is particularly hot or cold wait for the IdentiFINDER to adjust to the external temperatures. For further questions please contact technical support.
Our SMEs are standing by to assist with issues and repairs.
Was this helpful? Yes No
Keywords: calibration, fine gain, "extreme heat cold"